Message 2 of 6. Message 2 of This would suggest me that there is communication through the GPIB. Its actually a pretty poor example, I’m surprised its in the driver package like that. Will keep you updated if it works.
|Date Added:||20 June 2008|
|File Size:||42.64 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Agilent Analyzer – IEEE (GPIB) Driver for LabVIEW – National Instruments
This is providing the source voltages and also measuring through the SMUs perfectly. Transportability across different computer platforms 2. For Windows NT, change references from win95 to winnt.
Sampling on HP I’ll supply an example of sampling code I use later today. The only exceptions are that it does not have a soft front panel or a knowledge base file.
Help me using Agilent C driver with LabView
The only input commands I did were to set up teh channel definition and source setup. At the same time I get an error on the monitor of D00 I think there should be somewhere in the block diagrams of the HP B Example.
Do not 41156b the driver in the directory in which it was installed. Second, there is no knowledge base file, which is primarily a description of a VXI board.
Message 1 of 6. It includes a comprehensive on-line help file which complements the instrument manual. Message 6 of 6.
The SCPI commands look similar for both, but there are probably subtle differences that are the result of new features in the Agilent C. It is built on top of, and uses the services provided by VISA.
Agilent 4156 Analyzer
I managedto get communication between the equipment HP and LabView. Did you try other commands which take a longer time to execute? The default error message for error 6 indicates an abort in the GPIB communication. A small update on the setup definition I use: Its actually a pretty poor example, I’m surprised its in the driver package like that. If you do not have internet access, the drivers can be obtained by contacting your country support organization.
What exactly is stopping us from doing this? Will keep you updated if it works. I looked on the manual for this error and I cound not find it. I am not sure if the Instrument Driver for the B adds custom messages.
Moreover I cannot see the trace of the sweep I expected a trace of 10 MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface. Hey again, This is providing the source voltages and also measuring through the SMUs perfectly.
Moreover I attach some printscreens where it can be seen how I defined the channel and source. After detailing a bit more the process do you still think there might be a connection drop out between the GPIB interface and HP?
I am new to the forum and also new to LabView. I will keep you labviwe with the outcome. I have a very old VI with a very old driver where you can obviously not choose an SMU or supply any source but can only read. Here should be what you are looking labvieww. A higher level of support from Agilent Technologies for the compiled driver 3.